This is infrequently used content, but to expand accessibility to a wider audience we are using a machine for translation. Please excuse any unanticipated translation errors. -Thank you, Leica Geosystems.
Take a deeper look at our Leica DISTO™ D110 (E7100i)
User Manual Leica DISTO™ D110 (English)
User Manual Leica DISTO™ D110(Español)
Quick-Start Leica DISTO™ D110 (English only)
Leica DISTO™ D110(E7100i) CE conformity declaration
Leica DISTO™ + Bluetooth® – getting started on Android
Leica DISTO™ + Bluetooth® – getting started on iOS
Leica DISTO™ + Bluetooth® – getting started on Windows 10
Leica DISTO™ + Bluetooth® – getting started on Windows 8
Leica DISTO™ D110 Brochure (English)
Leica DISTO™ D110 aka E7100i Brochure (Español)
Leica DISTO™ Self-Calibration (English only)
How can I remove the pocket clip of the device?
Please proceed as follows to remove the pocket clip. Press the bottom of the device to unlock and remove the clip. Click here to purchase a replacement clip.
How can I attach the pocket clip to the device?
Please proceed as follows to attach the pocket clip. Please insert the clip through the gap in the bottom of the device.
How can I open the battery compartment?
In order to open the battery compartment on the pocket clip needs to be removed. After removing the clip, the battery compartment can be opened.
What is the maximum measuring range?
The device measures distances between 0.2 up to 60 meters.
What functions are implemented in the device?
You can measure distances, areas and execute continuous measurements. You can transfer all measurements via Bluetooth® Smart.
How can I turn the beep of the device on / off?
By pressing the Bluetooth® key and the area key together for 2 Sec the beep can be turned on / off.
How can I turn on/off Bluetooth® on the device?
By pressing the Bluetooth® key for 2 seconds.
How can I get one extra year of warranty for the Leica DISTO™ D110 (E7100i)?
Our standard warranty of 2 years can be extended by one additional year if you register your device here within eight weeks of the purchase date. If the product is not registered, a two year warranty applies.
What does it mean when I attempt a measurement but the unit shows a number on the display versus my measured value?
This is a message code. The numbers are displayed if a successful measurement could not be performed. The user himself can correct the measuring situation to achieve a measurement. Below you can find a table showing corrective actions:
|204||Calculation error||Perform measurement again.|
|240||Data transfer error||Repeat procedure.|
|252||Temperature too high||Let device cool down.|
|253||Temperature too low||Warm device up.|
|254||Battery voltage too low for measurements||Change batteries.|
|255||Received signal too weak, measuring time too long||
Change target surface (e.g. white paper).
|256||Received signal too high||Change target surface (e.g. white paper).|
|257||Too much background light||Shadow target area.|
|258||Measurement outside of measuring range||Correct range.|
|260||Laser beam interrupted||Repeat measurement.|
What does ISO 16331-1 standard mean?
When products and services meet our expectations, we tend to take this for granted and be unaware of the role of standards. However, when standards are absent, we soon notice. We care when products turn out to be of poor quality or are unreliable or dangerous. An ISO Standard ensures consistent quality and reliability for a product.
The range and accuracy of laser distance meters depend greatly on lighting conditions and the reflective properties of the target object. For Leica Geosystems it is important that the performance of the device is maintained not only in the test laboratory but even more so on everyday site tasks. Therefore we have worked with external experts to develop a global standard to test and specify laser distance meters and allow comparisons between devices of different manufacturers. Instruments tested in accordance with ISO 16331-1 achieve their promises. Further information can be found here on our website.
What Bluetooth® technology is integrated in the Leica DISTO™ D110 (E7100i)?
The Leica DISTO™ D110 (E7100i) has Bluetooth® SMART. The devices is working with smartphones or tablets with Bluetooth® 4.0, or are “Smart Ready”. It is also required that the operating system supports this Bluetooth® standard which is the case for iOS devices, Android devices 4.3 or higher, and Windows 8.0 or higher.
Can I send measuring data from a Leica DISTO™ D110 (E7100i) to iOS mobile devices?
Yes, the Leica DISTO™ D110 (E7100i) can communicate with iOS mobile devices. It’s possible to send measuring data to an iPhone (iPhone 4S, iPhone 5, iPhone 5s, iPhone 5c), iPad3, iPad 4, iPad mini, iPod touch (Gen. 5). (--> Bluetooth® 4.0)
Can I send measuring data from a Leica DISTO™ D110 (E7100i) to Android devices?
Yes, the Leica DISTO™ D110 (E7100i) can communicate also with Android devices, with operating system 4.3 or higher, and which have integrated Bluetooth 4.0. Used apps should support the corresponding Bluetooth® standard (e.g. Leica DISTO™ sketch).
Can I send measuring data from a Leica DISTO™ D110 (E7100i) to a Windows computer?
It is possible to send data to computers running with Windows 8.1 by using the free software DISTO transfer. This free software you can download from our DISTO webpage. Windows 7 and older Windows operating systems do not support Bluetooth 4.0 and therefore do not support data transfer from the Leica DISTO™ D110 (E7100i).
Can I send measuring data from a Leica DISTO™ D110 (E7100i) to Windows mobile, Windows RT or a Blackberry device?
No, this is not possible.
Does the Leica DISTO™ D110 (E7100i) offer apps?
Yes. We offer the Leica DISTO™ sketch app. The app is available free of charge for iOS devices on iTunes, and for Android devices on Google play.