3D DISTO User Manuals
Leica DISTO™ D2 User Manuals
Leica DISTO™ D2 User Manual (English PDF)
Leica DISTO™ D2 User Manual (PDF En Español)
Leica DISTO™ D2 CE Conformity Declaration
Leica DISTO™ D2 Recycling Passport
Leica DISTO™ D2 Brochure (English)
Leica DISTO™ D2 & E7100i Brochure (English)
Leica DISTO™ D2 & E7100i Brochure (Español)
Find out more about our Leica DISTO™ D3
How accurate is the tilt sensor in the Leica DISTO™ D3?
The accuracy of the tilt sensor is +/- 0.3° to the laser beam and +/-0.3° to the housing. The accuracy varies according to the measured angle and distance.
What units of tilt are integrated into the Leica DISTO™ D3?
You can set the Leica DISTO™ D3 to use different units of tilt. The user can select between ° and %. The units can be changed in menu item 2
Learn more about the Leica DISTO™ DXT
How do I change batteries of DISTO™ DXT?
Fold out the end-piece and open the locking mechanism with a coin or a screwdriver and lift the battery compartment cover. Insert batteries, observing the correct polarity. Close the battery compartment carefully. To ensure the instrument remains watertight, any dirt on the battery compartment seals should be avoided.
Take a deeper peek at our Leica DISTO™ D8
How many zoom levels does the Leica DISTO™ D8 have?
The integrated digital Pointfinder in the device has 1x, 2x and 4x zoom. You can see on the left up corner which zoom level you have selected. To change from one zoom level to the next one you have to press the “Digital Pointfinder” key.
Discover more about the DISTO D3a BT's predecessor.
How accurate is the tilt sensor in the Leica DISTO™ D3a?
The accuracy of the inclination sensor in the device is ± 0.3° to the laser beam respectively ± 0.3° to the housing of the device.
How can I check if the tilt sensor of my device is still within the described specifications (± 0.3°)?
The tilt sensor has to be switched on. On an absolute flat surface the first measurement has to be made. (please note angle a). Afterwards the device has to be turned around 180° and the second measurement has to be made. (please note angle b).
Now angle a and angle b are added up and divided by 2. If this value falls within ± 0.3° the tilt sensor of the device is within the described specification.
1 All about the DISTO™ A6 and its software DISTO™ transfer
Which system requirements does my PC need to work with DISTO™ transfer?
Generally the DISTO™ transfer works on all desktops, labtops, tablet PCs and UMPCs equiped with Microsoft Windows SE, ME, XP or Vista. Of course the computer must have a Bluetooth interface or you have to install a Bluetooth stick. Using an external Bluetooth stick the computer must fullfil the requirements of the Bluetooth stick.